By measuring the sample with both EDX and FTIR systems and using EDXIR-Analysis to analyze both EDX and FTIR data, elements can be identified automatically with high accuracy.
This package can be used to control g/g-level concentrations of 12 of the 24 elemental impurities specified in ICH Q3D Guideline for Elemental Impurities, including Class 1 elements Cd, Pb, As, and Hg, Class 2A elements V, Co, and Ni, and Class 2B elements Ir, Pt, Ru, Rh, and Pd.
(Option for EDX-7000)
Measure the Samples Kept in the Holder with EDX and FTIR
The Holder Can Be Used as the Sample Stocker after the Measurement
■ Vacuum Measurement Unit
Use this unit for sensitive measurements of light elements. It requires space for installation of a rotary pump and switch box at the side or rear of the desk supporting the main unit.
■ Turret Unit
Turret for 12 samples. It permitscontinuous measurements of samples up to 32 mm in diameter. It improves throughput, especially for measurements in a vacuum or helium atmosphere.
■ Mylar Film
Sample-holding film (for heavy element analysis)
■ Helium Purge Unit
This unit is used for highly sensitive measurements of light elements in liquid samples. Does not include a helium cylinder or regulator.
(Option for EDX-7000)
■ Screening Analysis Kits
RoHS/ELV Screening Analysis Kit
- With check samples for five elements RoHS and Halogen Screening Analysis Kit
- With check samples for six elements RoHS, Halogen, and Antimony Screening Analysis Kit
- With check samples for seven elements
■ Polypropylene Film
Sample-holding film (for light element analysis)
■ Sample Cells
3571 General Open-End X-Cell (no lid)
(Outer diameter: 31.6 mm, volume: 10 mL) Polyethylene sample cell for liquid and powder samples.
3529 General X-Cell (with lid)
(Outer diameter: 32 mm, volume: 8 mL) For liquid samples. Equipped with a relief hole and liquid retainer in case of liquid expansion.
3577 Micro X-Cell
(Outer diameter: 31.6 mm, volume: 0.5 mL) For trace samples. Recommended for use with a collimator.
3561 Universal X-Cell
(Outer diameter: 31.6 mm, volume: 8 mL) For liquid and thin-film samples. Equipped with a relief hole and liquid retainer in case of liquid expansion. Equipped with a ring to tightly hold thin-film samples with film.
|Measurement principle||X-ray fluorescence spectrometry|
|Measurement method||Energy dispersion|
|Target samples||Solids, liquids, powders|
|Measuring range||11Na to 92U (EDX-7000) 6C to 92U (EDX-8000)|
|Sample size||W 300 × D 275 × approx.H 100 mm (excluding radiuses)|
|Maximum sample mass||5kg (200g per sample when using turret, Gross mass 2.4kg)|
|X-ray tube||Rh target|
|Voltage||4 kV to 50 kV|
|Current||1 μA to 1000 μA|
|Cooling method||Air-cooled (with fan)|
|Irradiated area||Automatic switching in four stages: 1, 3, 5, and 10 mm diameter|
|Primary filters||Five types (six, including the open position), automatic replacement|
|Type||Silicon drift detector (SDD)|
|Liquid nitrogen||Not required (electronic cooling)|
|Measurement atmosphere||Air, vacuum*1, helium (He)*2|
|Sample replacement*||12-sample turret|
|Sample observations||Semiconductor camera|
|Memory||2 GB min. (32-bit), 4 GB min. (64-bit)|
|HDD||250 GB min.|
|Optical drive||Super multi drive|
|OS||Windows 7 (32-bit/64-bit)*3|
|Qualitative analysis||Measurement/analysis software|
|Quantitative analysis||Calibration curve method, correction for coexistent elements, FP method, film FP method, background FP method|
|Utilities||Automatic calibration functions (energy calibration, FWHM calibration)|
|Instrument status monitoring function|
|Analysis results tabulation function|
|Temperature||10 °C to 30 °C (temperature fluctuation rate 2 °C/hour max., temperature fluctuation range: 10 °C max.)|
|Relative humidity||40 % to 70 % (no condensation)|
|Power supply||100 V AC ±10 %, 15 A earthed socket|
|Dimensions||W 460 × D 590 × H 360 mm|
|Weight||Approx. 45 kg|
*1 Option for EDX-7000/8000
*2 Option for EDX-7000
*3 Microsoft Office is not included.
Vacuum measurement unit (optional) consists of a control switch box and rotary pump.
For Research Use Only. Not for use in diagnostic procedures.
This page may contain references to products that are not available in your country.
Please contact us to check the availability of these products in your country.